Newheek manufactures radiography table with four way table surface floating, which can be assembled with different brand and model of x ray tubes, such as Toshiba x ray tube. When install the x ray tube on Newheek radiography four way floating table, there are some attentions need the engineer to understand. Please see the following picture.
It shows the rotation electromagnet gap is large when installation. This problem not belong to the quality problem. But It needs the engineer to make adjust when installation according to the actual tube.
The adjust method as follows:
(1)loose the cap (see picture);
(2)then fasten the screw;
(3)then loose the screw for above 45° till when rotation, there is no resistance.
(4) fasten the cap.
Tel: +86 536-8882360 / +86 17616362243
Company: Weifang Newheek Electronic Technology Co., Ltd.